# PDUS210 – 210 Watt Ultrasonic Driver / Generator

The PDUS210 is a complete solution for driving precision and high-power ultrasonic actuators. Functions include high-speed resonance tracking of both series or parallel resonance modes, vibration amplitude control, and analysis functions such as impedance and frequency response measurement. The PDUS210 is well suited to both OEM product integration and laboratory use for research and development. Applications include ultrasonic drilling and cutting, medical devices, dental devices, ultrasonic testing, liquid cavitation, and vaporization.

The PDUS210 is controlled via USB and the included software package. An RS485 interface also provides a straight-forward method to control and monitor the amplifier for automatic test and OEM applications.

The PDUS210 generates a pure sine-wave output which avoids the excitation of secondary resonance modes by the drive harmonics. This makes it ideal for operating at the electrical parallel resonance, or ‘anti-resonance’. This operating point is close to the mechanical resonance frequency but is less sensitive to changes in load dissipation, which is useful in precision machining applications where constant vibration amplitude is desired.

The PDUS210 is available with standard output voltage ranges from 17 Vrms to 282 Vrms, and current ranges from 0.7 Arms to 11 Arms. These ranges are optimized for load impedances ranging from 1.5 Ohms to above 400 Ohms at resonance. For research and development applications, a reconfigurable version is available (PDUS210-FLEX), this version uses external output matching transformers to allow operation at any of the available output voltage ranges.

Specifications and User Manual

#### Mechanical Specifications

 Specification Value Notes Enclosure Dimensions 227 x 168 x 54 mm L x W x H Mass 1.4 kg Temperature Range 0C – 50C Humidity Non-condensing

### PDUS210-FLEX Specifications

The PDUS210-FLEX is identical to the standard PDUS210 except that it requires an external transformer connected between the amplifier and transducer using the supplied cable. This allows the user to switch between different output voltage ranges by changing the external transformer. Please refer to the manual for unipolar output voltage options and operating instructions.

The PDUS210-FLEX must be purchased with at least one external transformer. These are available online. The part numbers and specifications are:

 Order Code Turns Ratio Voltage pk-pk Voltage RMS Amps pk-pk Amps RMS Optimal Load Ohms Load Range Ohms* Frequency kHz TX210-800 18.18 800 282 2 0.71 400 260 – 840 20-200 TX210-600 13.64 600 212 2.6 0.92 225 146 – 472 20-200 TX210-400 9.09 400 141 4 1.4 100 65 – 210 20-200 TX210-200 4.55 200 70 8 2.8 25 16 – 52 20-200 TX210-175 3.98 175 62 9.1 3.2 19.1 12 – 40 10-100 TX210-100 2.27 100 35 16 5.7 6.25 4 – 13 20-200 TX210-50 1.14 50 17 32 11.3 1.56 1 – 3 20-200

A kit containing five transformers ranging from 35Vrms to 282Vrms is also available (TX210-Kit1). This includes the following output voltage ranges: 35, 70, 141, 212, and 282 Vrms.

 Specification Value Notes Input Connector Plug-in screw terminal Connecting cable supplied Output Connectors Identical to PDUS210 See “Front Panel” section Transformer Dimensions 104 x 57 x 51 mm L x W x H Mass 0.2 kg

### Desktop Software

The desktop application controls the amplifier and plots the key signals and operating conditions, which can be logged and exported as CSV files. The desktop application also includes advanced analysis functions such as impedance frequency response analysis, and a setup assistant to help new users analyze and choose operating conditions. Please refer to the manual for a full description.

### Standard Delivery Contents

• PDUS210 Amplifier (in chosen configuration)
• Handheld controller
• Handheld controller cable (DSUB9 3ft)
• 280W Desktop power supply
•  IEC Power cable suited to the destination shipping address
• USB Cable (Type A)
• 3 Way plug-in screw terminals for output (Amphenol TJ0331530000G)
• 4 Way plug-in screw terminal for RS-485 connector (Amphenol TJ0431530000G)

### Warranty and Service

The PDUS210 is guaranteed against manufacturing defects for 12 months from the date of purchase.
Contact your distributor or info@piezodrive.com for service. Please include the amplifier serial number.